Laser Machine Physiotherapie, Infrared Heating Plus Electrode Pads
Infrared Therapy Apparatus
I. Product Introduction of the infrared heating plus electrode
Brief introduction: intelligent pain therapy is operated by a micro-computer, which changes electricity into light that can be applied to body surface irradiation. It is mainly used for in pain adjuvant therapy of soft tissue.
II. Main Characteristics and Requirements
1. Working environment requirements:
2. Power requirements:
a) temperature: -10ºC~+40ºC;
b) relative humidity: 30%~75%;
c) atmospheric pressure: 700hPa~1060hPa;
3. Power requirements:
a) power AC220V, frequency 50Hz.
b)rated power input: 80VA
4. Two channels for output: dotted and planar
5. Dotted radiator, length: 160mm±5mm, OD: 30mm±2mm
6. Dotted output wavelength: 810nm±5%.
7. Dotted output power: 0~500mw, range: 1mw.
8. Planar radiator, length×width×ply: 200mm×160mm×50mm.
9. Planar output wavelength: 980/810nm±5%
Wavelength at 980nm, adjustable power output: 1~200mw, range 1mw;
Wavelength at 810nm, adjustable power output: 0~500mw, range 1mw.
10. Treatment time: adjustable between 0-90min, range 1min.
11. Size(length×width×ply): 450mm×380mm×780mm;
III. Structure and Components of the Product
The product is composed by the main engine, the bracket and the radiator(dotted radiator and planar radiator), etc.
IV. Range of Application
This equipment is mainly used for adjuvant therapy of soft tissue pain.
This equipment should not be applied with the following diseases and body parts: Acute stage of inflammation, febricity, progressive wasting diseases, noninflammatory edema, body parts with high organic circulation disorder, body parts with high perceptual deficits, eye area, thyroid area, diseases which may cause internal bleeding.
VI. Notes, Cautions and Indications for Use
1. During treatment, avoid direct irradiation of eyes or genitals.
2. Do not look directly to the radiators.
3. Read this manual carefully before application in order to ensure safe operation, correct installation and efficient maintenance.
4. Applying the equipment under doctor's guidance.
5. The application on children, elders, patients with acute diseases and patients with no self-care ability should be monitored by physicians.
6. During application, the nominal ocular hazard distance(NOHD) is 4.90m. Please pay attention to the safety distance and avoid direct irradiation of unaffected area or of eyes, otherwise it will cause irresistible damage to the users' eyes.
7. Before application, check if the single phase three wire system power supply used matches with what the nameplate requires. Make sure the integrity of the earth plug; If not, please add extra earth plug to the ground terminal to ensure safety.
8. It is absolutely forbidden to cover the ventilation slots during treatment. The equipment should be positioned 200mm or farther from walls to ensure its ventilation and heat dissipation. Keep away from heating device.
9. Do not use the equipment in environment of high humidity. Make sure the power plug is out of socket during cleaning. Liquid and other matters should not get in the inside the equipment, otherwise it might lead to electric shock or device damages.
10. Must avoid using flammable anesthetic or oxidizing gas such as nitrous oxide(N2O), oxygen. Do not insert the optical fiber into the disinfectant directly. Wipe the surface with disinfectant. Making the disinfection solvent and combustible solution volatile before using them., Should pay attention to the risk of internal gas ignited.
11. Before turning on the power, please make sure if all the fittings is connected firmly. If not connected inside short chain remote plug, therapeutic apparatus will be stopped from output.
12. Long time using the apparatus,should pay attention to heat dissipation.
13. Using this apparatus, should keep it far away from the interference sources of electromagnetic radiation, such as short wave, microwave, CT, Nuclear magnetic and other high frequency equipment. Usually stay away from more than 30 meters and used across the room or shielding. Do not use one power supply with others. Otherwise may cause unstable output.
14. When the intelligent pain therapy apparatus is working, do not plug or unplug the laser plugs, Otherwise it will damage the radiator, And can cause radiation damage to people. When turning off the apparatus, turn off the key switch first,then the machine automatically enter the shutdown procedure. After executing the shutdown procedures, Will automatically shut off the power. Finally pull the plug. When the machine performs shutdown procedure, do not unplug the power plug directly, so as not to damage the machine.
15. If there is an accident during the operation process need to stop working,please immediately put the emergency stop button in blocking state to stop the output.If need to use again, please gently rotate in the label direction,the button will lift automatically,then will exit the blocking state.Do not twist the wrong direction.
16. During the specific operation process, should be combined with the patients' specific situation and doctor's experience to use the apparatus flexibly.
17. After using this apparatus,should take down the key,in order to avoid non-work personnel making unsafe operation.
VIII. Installation Method
1. Avoid forced unpacking, protect the package and the equipment.
2. Fix the probe units and the fix screws of brackets hand wheel.
3. Raise the bracket by loosing the hand wheel(Note: Conduct this movement slowly in case the bracket hit the user).
4. Rotate the bracket unit clockwise. Push the bracket to proper position and then fix the fix screw of the hand wheel.
5. Install the radiators on the right side of the equipment otherwise the power output will be influenced.
6. It just need to connect power to start.
Note: Ensure the function of earth plug before connect power to the equipment.
IX. Instructions for use
9.1 the information for clinic use
Selective irradiation dose
The irradiation dose is radiation energy, varying with the medical record.
The irradiation dose is the product of the radiation power and duration
The radiation power
Two dot output, each radiation power is 0 ~ 500mW
select the duration
Contact exposure, generally for3 minutes. Noncontact irradiation (light end from the wound in 1 ~ 2 cm) extend the time. If according to the conventional irradiation for 3 to 5 minutes the effect not beautiful, may be appropriate to extend the exposure time is 6 ~ 10 minutes.
How to output
when start the Emission output, don't regulate the increase and decrease, to protect the mis-regulate. when stop the Emission output, it can regulate the increase and decrease.
Press the start key , the light of stop is off. Said into the output ready state.after 2 seconds of delay, it begin to emission.
Just press the start button (START) key, continuous output mode, the logo is (START) lamp, lamp (STOP)
At the same time according to the key (START) and (STOP)key, pulse output, the logo is (STOP) and (START) light frequency of 10Hz lamp, said transmitter output periodic pulse, frequency of 10Hz.
Acupoint irradiation should as far as possible in accordance with the Chinese medicine dialectical acupoint and acupoint principle.
The specific operation, should be combined with the specific condition of patient and doctor's experience, flexible application
9.2 Treat operation
The three core 220V power use of the instrument (i.e.,power supply must be protected earth wire), to confirm the power correctly, can insert the power plug into the power socket connected to power.
2. start the device
The key is inserted into the "power switch", clockwise rotation of 90 degrees, the green power indicator light(POWER). That the power supply is switched on. According to the "standby / ready" button, the yellow button indicating lamp is bright, enters the ready state. Then the machine can start treatment. Click the "standby / ready", enter the standby state, not the output laser.
3. set up the left\right emission
By pressing the left \right selection key, can switch between the left and the right way to lose.
According to the "digital select" button, you can switch between time and power setting.
4. treatment time set up the power
Select either the left or right of the laser, and then select themodify time or power, the screen is selected item numbers flashing, that can be adjusted. According to the disease,according to increase or decrease the key, can set or modify the required treatment time and power.
Press the"increase"key and the "decrease"key,, Adjustable parameter value, Adjustable power range of 1 to 500 MW, 1 MW step.. Working time in 1 to 99 minutes between continuous adjustable, step 1 minutes.
If the button is pressed continuously, rapidly increasing or decreasing. Output in the process, can increase or decrease the power.
5. start and stop
On the left in the laser setup is complete, you can start the left laser treatment, according to the left to start (START)key, 2 second delay, continuous mode emission laser therapeutic instrument; at the same time according to the left to start (START) key and stop (STOP) key, transmitting laser therapeutic instrument left pulse; such as the need to stop treating according to the left. Stop (STOP) key, can stop the treatment.
In the right laser setup is complete, you can start the right laser treatment, according to the right start (START) key, 2 second delay, continuous mode emission laser therapeutic instrument; at the same time according to the right (START)key and start stop (STOP) key, mode emission laser therapeutic instrument right pulse; for a half-way stop treatment, according to the right stop (STOP) key, can stop the treatment.
Laser output, in 1 minute increments the countdown. Gaugeto zero, stop work, return the set value, and issued 8seconds alarm.
Arbitrary and emergency, can stop the output, according to the "emergency stop" button. According to the "emergency stop" button icons, clockwise rotation, you can reset, press the "standby / ready" button, you can start again.
Punctate radiator installation guide bar can contact irradiation.
After the end of the work, cut off the power supply, remove the power plug from the "socket". During removal, the need for walking plug, not pull pull cable.
1. The room should be kept clean, with no-corrosive gases, none interference of strong magnetic field and electric field.
2. The intelligent instrument for pain treatment is optical equipment. Be careful not to bump when installing and using the device, handle with care and no disassemble. Disassemble will have a strong impact on the output of optical power.
3. In order to avoid cross infection and maintaining the effective use of therapeutic equipment. The main light guide stick of laser head should be disinfection by alcohol swab and be properly kept after treatment.
4. Power cord should not have excessive bending, stretching, twisting and binding. It should not bear heavy pressure so as not to damage power cord.
5. Keep device clean and pay attention to moisture, dust-proof issue. Turn off the power switch, unplug the power cord, covered with protective cloth when long time no use. Don't place heavy objects on the device.
6. Long time without use the equipment, power up to check the device once every six moths.
7. The device should return to manufacture to maintenance to ensure stable performance and continue to use.
8. Replace the fuse plug.
-remove the general supply plug from the socket
-use slotted screwdriver to lift under the fuse cover.
-take away the burned-out fuse.
-install the eligible fuse
-cover the fuse lid
-see fuse models on the identification around the fuse base.
-make call to professional maintenance person when the device break down. Contact with our dealer or services department of our company when you require service of our company. Avoid arbitrarily repairing.
XI. Transport and storage requirements
1.Transport and storage conditions
Environment temperature -10ºC~55ºC
Atmosphere pressure 500hPa~1060hPa
2. Transportation:Please handle with care; Never put it upside down; Protect it from rain or snow.
3. Storage: Packaged traction bed shall be stored in room with good ventilation, no corrosive gas, having a relative humidity lower than 80%.
XII. Production date and validity period
Refer to the qualification certificate for production date
Validity period is 6 years
XIII. Environment protection
At the end of its life span, or when it is worn out, the worn parts shall be dealt according to local relative environment protection laws. Never throw it away carelessly.
XIV. Circuit diagram and electronic components list
We can provide circuit diagram and electronic components, electronic components list, figure legends, correction details and other technical data when required.
XV. Simple troubleshooting
|No display after starting||Bad connect of power socket||Check the socket|
Fuse broken through
|Bad connect of output line|
Check output line socket
It the problem still can not be solved, please contact the manufacturer by the contact data on warranty card. Never open the host for fear of damaging other parts.
XVI. Accessories list (refer to packing list for details)
|1||Host Machine||1 set|
|2||User Manual|| 1 copy|
|3||Qualification certificate||1 copy|
|4||Warranty Card||1 copy|
Welcome to contact us for the infrared heating plus electrode.