Magnetic Andullation Therapy, Transcranial Agnetic Stimulator Uni
Transcranial Magnetic Stimulation Apparatus
Technical features of the magnetic andullation therapy:
2 channels magnetic therapy, 2 channels FNS(Fastigial Nucleus Stimulation) and 4 channels limbs electrical stimulation output
Magnetic field output parameters of the magnetic andullation therapy:
1. Frequency conversion magnetic field sensor frequency F1-F6 adjustable (5Hz ,10Hz, 20Hz,30Hz,40Hz,50Hz), convert frequency automatical, only need 10s±1s per conver
2. Constant frequency model available as well. Any frequency (F1-F6) could be selected.
3. 2 gears available. strong gear:13mT-25mT; weak gear: 3mT-13mT
4. treatment time:20min-30min
5. there are 9 generator on the magnetic cap responding to 9 acupoints, which have vibration function.
FNS parameters of the magnetic andullation therapy:
1.output wave: continuous wave, dilatational wave, light lamented wave, massage wave E1, massage wave E2, massage wave E3
2. output pluse strengh :0~42Vp-p±10%, 0~99 adjustable
3. FNS treatment time: 20min
Limbs electrical stimulation parameters:
1. working frequency:2KHz~6KHz
2. modulation frequency:0~150Hz
3. modulation wave: sine wave, square wave, triangle wave,index wave, sawtooth wave
5. less than 100mA, 0~99 adjustable
6. heated electrode plate temperature: 38ºC~55ºC
7. alarm indication when the treatment is completed.
Welcome to contact us for the magnetic andullation therapy.