Low-Intensity Portable X-Ray Imaging Scope
Low-Intensity Portable X-Ray Imaging Scope (the following in short X-Ray Imaging Scope), a new type of X-Ray medical instruments in the world, consists of main engine and power supply. From the appearance, the main engine adopts the structure of C-Type-Arm, which amplifies the radius of observation. At the same time, the materials select ABS engineering plastics, heightening the insulation capacity.
Because of adopting MCP-X-Ray imagine strengthening device, having compact texture, high gain, high resolution, the scope can produce clear real-time images without darkroom. The advance technology of widening pulse is used to control electrical appliance of X-Ray tube, which has many measures to be protected, so it has high working stability and no noises.
In addition, the scope takes strict precautions, the leak of X-Ray reducing, and then the operators do not need any protections. This scope applies to orthopedics and pediatrics in the hospitals and to diagnose IUD for childbearing-age women in time. Especially, it applies to various field rescues in such as industry materials and apparatus quality inspection, athletes, armies, field working and safety checking, and so on.
The scope belongs toIClass B Model Equipment.
Specifications and Conditions
Output image size: Φ50mm/Φ75mm
Survey thickness: ≤300mm
X-Ray tube voltage: 35~70kv/90kv continual adjust
Voltage accuracy: ≤±1%
Tube current: 0.25~0.5mA continual adjust
Current accuracy: ≤±2%
Ratio : ≥4%
Gray level: ≥10 level
Main engine working frequency: 20KHz
Radiation leakage: the corresponding National Standard GB9706.12
Opposite humidity: ≤80%
3. Power supply
Power consumption: 80W
Resistance maximum: 0.5
Releasing current machine: 1 A
4. Shape size and weight
Length × width × thickness: 620×500×190
Main engine: 3.6Kg Guard: 0.61 Kg
Gross weight: 11Kg
X-Ray Imaging Scope is made up of different parts in the pictureI. When put through power supply, turn on "start" button, and then the scope begins to work. The pulse signal from main engineIamplified through efficiency reaches to X-Ray tube anode, while the pulse signal from main engineII amplified reaches to X-Ray tube wick, which releases X-Ray. Then the corresponding number KV/μA will come out on the board. This moment the object is laid between X-Ray source and image strengthening machine, on whose monitor we will see the clear image. In order to work stability, the system adopting the technology of widening pulse, makes tube current and tube voltage stable, and the X-Ray tube works in good condition. Due to the function of high voltage starting slowly, the X-Ray tube anode doesn't have the phenomenon of high voltage being too high. The main control machine with the mini-slice equipment working at the frequency of the 20KHz, improves the entire system's efficiency and clears away the noises, and it supplies workers for quite environment with a smaller volume. The power supply of perspective device adopts a switch with high frequency and high efficiency, having overall protections. To ensure the safety of perspective device, the scope takes different measures, so the device is safe and reliable.