Standard Class B Dental Autoclave 16L
|FOB Price:||US $1-999 / Piece|
|Min. Order:||1 Piece|
|Min. Order||FOB Price|
|1 Piece||US $1-999/ Piece|
|Production Capacity:||10000 Sets / Year|
|Payment Terms:||L/C, T/T, Western Union, Paypal, Money Gram, Cash|
- Model NO.: MC-KM16
- Classification: Class B
- Mains Fuses: F 15A
- Test Programmers: Helix/Bd Test; Vacuum Test;
- Pre-Heating Time(From Cold): Approx 20 Mins
- Package Dimensions(L*D*H): 755*584*532 mm
- Trademark: Med Equipment
- Origin: Guangzhou China
- Certification: ISO, CE
- Insulation Class: Class II
- Noise Level: 50 dB(a)
- Working Temperature: 121ºC-134ºC
- Extenrnal Dimensions(L*D*H): 625*425*456 mm
- Capacity of The Distilled Water Reservoi: Approx. 5.0 L (Water at Level Max)
- Transport Package: Standard Packing for Export
1.Standard Class B with three times vacuum and drying, the remaining temperature of sterilized instrument is less than 0.2%.
2.The vacuum data can reach -0.8bar,it fits various instrument,including package,unpackaged,solid,hollow,multi-apertures and inset pipes. It makes sure the instrument can be sterilized completely.
3.Fuzzily computer controlled, digital bright show. The interface is modularized panel for easy operation.
4.It is installed with BOWIE&DICK which measures the penetration of water steam.
5.it has vacuum text which can make sure the ability of texting vacuuming.
6.Sterilization cycle self-comment.
7.Fast and independent steam generator.
8.Large capacity instrument tray made of SS. SU304.
9.Safety lock and triple temperature.overload protecting
10.Auto/Manual water adding
11.Inset cleaning device
12.Special function of creating medicine cotton and sterilizing rubber items
13.Handsome aspect and good quality
14.Cristal LCD display
15.5-Button panel with the LCD display, so convenient.
16.Warning message, Error is made easier to judge.
- Please place the sterilization on a plane place , which could adjust the height by adjusting foot pad
- Please adopt distilled water to extend the service life
- Don't block the heat removal system on the surface of sterilization
- Please put the instruments on the tray not touching one another allowing space for steam to pass through
- Please empty the used water tank frequently. Usually it need empty when the clean water in the reservoir is empty.
- Please ensure the door handle is closed tightly
- Never try to open the chamber door if and when the pressure display doesn't show "0.0"
- Please cut off the power and cool down the auto clave before maintenance
- Don't drag the machine while moving
- Power connection must be grounded properly
- Must be provided enough power. (23L:2300VA/18L:1800VA)
- While the outside temperature is too low, please pre-heating the machine without instruments for 30 minutes
- Don't cut off the power to stop the machine directly. If you need stop the machine directly, please press the
"START/STOP" button for 3 seconds.
|Model (LCD/ LED)||16 L||18 L||23 L|
|Insulation Class||Class II|
|AC Power||220V/50Hz; 220V160Hz; 115V/60Hz;|
|Mains Fuses||F 15A|
|Noise Level||50 dB(A)|
|Test Programmers||Helix/BD Test; Vacuum Test;|
|Pre-Heating Time(From cold)||Approx 20 mins|
|Net /Gross Weight||39KG/45KG||43KG/49KG||45KG/52KG|
|Chamber Dimensions||230*360 mm||250*380 mm||250*380 mm|
|Extenrnal Dimensions(L*D*H)||625*425*456 mm|
|Package Dimensions(L*D*H)||755*584*532 mm|
|Operating Environment||Temperature: 5ºC-40ºC|
Relative Humidity: max 80%,
Height: max 3000 m (a.s.l.)
|Capacity of the Distilled Water Reservoir||Approx. 5.0 L (water at level MAX)|