Med-Ob-Fxq-3 Infant Incubator
|FOB Price:||Get Latest Price|
|Min. Order:||1 Piece|
|Production Capacity:||10000 Sets/Year|
|Transport Package:||Standard Packing for Export|
|Payment Terms:||T/T, cash|
- Model NO.: MED-OB-FXQ-3
- Certification: CE, ISO13485
- Constant Temperature Time: Not Less Than 2h
- Trademark: Med Equipment
- Origin: Guangdong, China (Mainland)
- Type: Infant Incubator
- Atmospheric Pressure: 70~106kpa
- Operating Time of Manual Temp-Control: Set as 15min
- Specification: CE, ISO
The product will be guaranteed freely for one year as from the date of purchase, except for any one of the following circumstances:
a) Wrong operation
b) Damage due to failure to follow the cleaning and maintenance method specified in the instructions;
c) Change or use any part that is not designated by the company;
d) Insert any accessory or device that is not in compliance with the safety requirement of the pro duct, causing any change in safety performance, product performance and parameters;
e) Due to external factors of force majeure (such as disaster);
The maximum load of emergency platform is 10N for the infusion stand, 20N for tray and 10Kg for the baby bed.
If not in use, the emergency platform should be kept indoors, with a temperature of -40~55 ºC, relative humidity of less than 80%, without any vibration and corrosion and with excellent ventilation,
In use, the emergency platform should be kept vertical with the ground, with an angle of less than 5° in case of being inclined.
The scale on the stop board of the baby bed is only for reference.
This model is not equipped with humidity control device and oxygen device.
Please use the accessories and parts of the factory; otherwise, the user shall be liable for any consequence.
With the emergency platform in use, its truckles should be locked to any shift during operation.
The temperature parameters of the emergency platform have been set in the factor.
2.1 Main Use and Applicable Scope:
The product is applicable for the body temperature maintaining, emergency operation and intensive care of new-born baby, sick and weak infant and premature infant.
2.2 Normal Working Environment
a) Power voltage: (220±22)V (50±1)Hz;
b) Ambient temperature: 18~30ºC, with the control temperature being at least 3ºC higher than the ambient temperature
2.3 Product composition: temperature controller, radiation source, bed frame, spot light and frame, etc.
2.4 Heater: radiation-type far IR heating element.
2.5 Normal Working Environment:
a) Ambient temperature: 18~30ºC, with the control temperature being at least 3ºC higher than the ambient temperature;
b) Relative humidity: 30~75%;
c) Atmospheric pressure: 70~106Kpa;
d) Ambient air speed: < 0.3 m/s,
e) Placed in an arena with relatively static air flow and without direct sunlight and other heat sources.
f) Power voltage: 220±22V, 50±1Hz
g) Consumed power: ≤1000VA
3.6 Basic Parameters
1) Illumination of spot light at the bed center: ≥100lx
2) Constant temperature time: not less than 2h
3) Temperature uniformity: ±2ºC with the baby bed at the horizontal level:
Mode for infant temp-control: alarming when the temperature deviating by +1.5ºC and -2.0ºC after constant temperature is set.
Manual temp-control mode: the heater outputs at fixed energy level and at max output energy level.
5) Measuring range of skin temperature: 25.0ºC~42.0ºC
6) Setting range of skin temperature: 25.0ºC~37.0ºC
7) Operating time of manual temp-control: set as 15min
3 Failure Analysis and Elimination
|No display and no alarming||Power switch is not on||Switch on the power|
|Power-off alarm light is on, sounding||Power is off||Switch off the power|
|Power cable is not connected||Connect the power cable|
|Fuse tube is damaged||Change the fuse tube and request for maintenance|
|Skin temperature sensor alarms||1. Skin temperature sensor is not connected||Connect properly the skin temperature sensor|
|2. The plug of the skin temperature sensor in the unit is off or improperly connected and damaged||Request for maintenance|
|3. Skin temperature sensor is damaged||Change the skin temperature sensor|
|Over-temperature alarms||1. Temperature control system fails||Request for maintenance|
|2. Independent over-temperature sensor is damaged||Request for maintenance|
|Skin temperature deviation alarms||1. Ambient temperature changes much||Control the ambient temperature|
|2. Skin temperature sensor is not in accurate contact with the infant||Observe carefully|
|3. Skin temperature sensor is damaged||Request for maintenance|
|Panel operating key fails||1. Penal plug is improperly contacted||Request for maintenance|
|2. Penal key is damaged||Request for maintenance|
|Displayer fails to display||Request for maintenance|
|No heating||Request for maintenance|
10 Packing List
|2||Skin temperature sensor||Set||1||In the document bag|
|3||Power cable||Pc||1||In the document bag|
|4||Infant bed cover||Pc||1||On the infant bed|
|5||Infant mattress||Pc||1||In the infant bed cover|
|6||User's Book||Copy||1||In the document bag|
|7||Fuse φ5×20/2A /0.2A||Pc||2 for each||In the document bag|