Have Stock in Guangzhou High Frequency Ventilator Respirator
|FOB Price:||Get Latest Price|
|Min. Order:||1 Piece|
|Transport Package:||Standard Packing for Export|
|Payment Terms:||L/C, T/T, Paypal, Money Gram, Western Union, Cash|
- Model NO.: MED-V-300JII
- Kind: Gynaecology
- Ventilator: Frequency:10~150 Breaths/Min
- Respirators: Sigh:6breaths/Min
- High Frequency Emergency Ventilator: Negative Pressure Suction:-0.07~0MPa
- Respirador Artificial: Volume Cycling Ventilation:T=10min, I/E=1:1.5
- Specification: CE, ISO
- Type: Ventilator
- Certification: CE, ISO13485
- Ventilator Respirator: I/E Ratio:1:1~1:3 and Inverse Ratio
- High Frequency Ventilator: CPAP:0.1~0.8kpa
- Ventilation System: Driving Pressure:0~0.15MPa
- Trademark: Med Equipment
- Origin: Guangzhou, China
High Frequency Emergency Ventilator (without Air Compressor)
It is widely used for clinical salvaging and resuscitate, also is a professional device used for
respiration salvaging on emergency ambulance.
Features of the ventilator respirator:
A. An opening system is designed to avoid leakage of airway, When ventilating, the intubation may have no
casing bag. Little tube may be used for ventilation when resistance of expiration is small, so that mucosa won't be
B. Airway is open during wentilation, suction and ventilation can be done synchronously, so good effectiveness
of oxygen combination can be shown without inspiration and expiration synchronizing.
C. Lower airway pressure is good to increase blood discharge and decrease cerebra pressure.
D. Suction device, its driving negative pressure is adjustable.
E. Storage battery can be used continuously for four hours after full charged.
F. The system disposes completely with laryngoscope, inlet intubation, catheter of tracheotomy, nasal prongs
tube, and air tube etc.
Specifications of the ventilator respirator:
I/E ratio: 1:1~1:3 and inverse ratio
Negative pressure suction :-0.07~0MPa
Driving Pressure : 0~0.15MPa
Volume cycling ventilation : T=10min, I/E=1:1.5
Adults:T1(100bpm):T2(20bpm)=7:3, Regular frequency : tidal volume≥400ml/breath, Driving pressure:0.06MPa
Children:T1(100bpm): T2 (30bpm)=7:3, Regular frequency : tidal volume≥300ml/breath, Driving
Storage battery:4Ah, 12V ; Rechargeable.
Maximum Output Oxygen Concentration :85%
Alarm:Alarm with light and sound when lack of power, or no air supply.
Ventilation Mode:HFJV(High Frequency Jet Ventilation);CPAP(Continuous positive airway
pressure),HFJV+SIGH;Volume cycling ventilation.
Welcome to contact us for the ventilator respirator.