Med-R-Xy-24 Model Relax Massager Medical Equipment
|FOB Price:||Get Latest Price|
|Min. Order:||1 Piece|
|Production Capacity:||10000 Sets/Year|
|Payment Terms:||T/T, D/P, Paypal, Money Gram, Western Union|
- Model NO.: MED-R-XY-24
- Environment Temperature: +5~+40oc.
- FOB Shanghai: $128
- Package: Carton Box
- Power: AC220V±22V; 50Hz±1Hz
- Classification: Physiological Functions of Diagnosis and Monitoring Equipment
- Trademark: Med Equipment
- Origin: Guangdong, China (Mainland)
- Type: Model Relax Massager
- Relative Humidity: ≤85%.
- MOQ: 10
- Atmospheric Pressure: 860hpa~1060hpa.
- Overall Dimensions Cm: 78×48×0~125
- Certification: CE, ISO13485
- Specification: CE, ISO
1. 1 Scope of main applications
The product is suitable for vibratory massage aimed at relaxing muscles of the whole body, and releasing muscle tension and weariness.
1. 2 The product standard being abided by is enterprise standard.
1. 3 Normal working conditions
2. basic parameters
2. 1 Overall dimensions (Length×Width×Height)cm: 78×48×90~125
2.2 Height adjustment gears: 8 gears.
3. 1 Stability: the massager can stand still on even ground without obvious sway.
3. 2 The surfaces within the massager's approach shall be smooth, having no burr, acute angle, or obvious injure.
3. 3 The height of massager can be adjusted appropriately within the range stipulated in 2. 2 Basic parameters.
3. 4 The massage positions are adjustable: you can bind the massage bondages as needed in training to lumbar vertebra, cervical vertebra, waist, buttocks, or legs, etc.
4. How to use
4. 1 You can adjust the height of control panel according to the need of training.
4. 2 Insert the power pin into power socket.
4. 3 Let the trainee stand on the fixed plate, and put the massage bondages on the correct part.
4. 4 Open the power switch, and begin relax massage exercise; at the end of exercise, just turn off the power switch.
5. Matters need attention
5. 1 The trainee shall read the user manual carefully to get familiar with the properties, requirements, and operating methods of the massager before using it.
5. 2 The trainee shall take exercise with the help of physical therapist or nursing personnel, following the principle of "Step by Step" to increase the level of training difficulty and training time gradually according to the trainee's condition.
5. 3 Power socket without ground wire shall not be used.
5. 4 The adjusting bolt must be firmly locked into the screw hole in the process of height adjusting.
5. 5 The service life of this device is 10 years (quick-wear parts are exceptional).